Olympus GIF-H260 Electronic Gastroscope.
1, The image is currently okay, the objective lens has wear and tear, Severe water ingress into the lens, CCD has hidden problems;
2, 20% of wires inside the light beam were broken;
3, The insertion tube has scratches and wear;
4, The angle is not enough, wire, bending section, spiral tube are worn;
5, Biopsy channel was broken and has serious air leakage;
6, Bending rubber is worn.
Replacement: objective lens, beam, insertion tube, bending section, spiral tube, wire, biopsy channel, bending rubber (image not warranted).
5-8 working days.
Do not place the endoscope together with other instruments and avoid touching other parts with the insertion part.
Wait until the endoscope angle returns to the 0-degree state before pulling out.
Since the insertion part is long, please be careful when cleaning and storing it, and be sure to confirm that it is not stuck beforehand.
When sterilizing EVIS endoscopes with ethylene oxide, remove the waterproof cap; when sterilizing other endoscopes with ethylene oxide, put on the ETO cap.
Wait until the endoscope angle returns to the 0-degree state, then pull out from the trocar needle and trocar.
When adjusting the angle of the insertion part, the angle should not be smaller than the range indicated in the instruction manual.
Please keep the sheath as straight as possible during operation.
Hold the insert gently and handle it carefully when taking it out.
Store in a dry and well-ventilated place at room temperature and in a clean condition.