From a simple bending rubber angulation adjustment to a complex image bundle replacement, total scope is the perfect choice for the flexible endoscope repair service. Our capabilities extend across all makes and models, allowing us to provide the convenience of a "one-stop shop" for our clients. We repair flexible endoscopes on-site in our state-of-the-art service center. Our technicians are professionally trained and have years of experience executing both major and minor repairs on all types of flexible endoscopes.
We provide proven flexible endoscope repair service to ensure your flexible endoscopes are patient-ready. All repair services include:
Distal Tip Replacement
Light Beam Replacement
Insertion tube Replacement
Light Pipe Replacement
Coil Pipe Replacement
Water container Replacement
Angulation wire assemble Adjustment and Replacement
Bending Rubber Replacement
FY MED has done a number of flexible endoscope repair and maintenance for our customers, and we are happy to show some cases to let viewers know about professionalism. Click here to see our Maintenance Case for Olympus GIF-H260 Electronic Gastroscope.
Bed side cleaning:
Dip in with soft cloth first enzyme wash clean soft outside surface of the lens, enzyme wash pipe cavity, after several times, and then into the discharge air enzyme wash, and then use leak-checking and observe hand pump for several seconds, a pointer is stable, as is normal, can undertake decontamination, if under dry conditions, please clean within 3 hours after using, in order to avoid dirt, dry, affect the quality of decontamination.
Please separate the Flexible Endoscope from other equipment during transportation to avoid damage.
Link the leak detector, close the pressure release button, squeeze the hand pump, hit the scale within the green range, and wait for several seconds, if the pointer becomes stable, the Endoscope is completely immersed in water, pay attention to the leak detector can not be immersed in water. Turn the direction, observe for 30 seconds, confirm that there is no continuous bubble overflow, first take out the Flexible Endoscope from the birdbath, then press the pressure release button, release the air, and go to the leak detector, leak detector is finished. Take care to dry the leak detector thoroughly.
Note: If the pointer continues to drop, the medical endoscope for sale may be seriously leaking or the leak detector may be damaged. Stop the leak detector immediately to avoid water entering the Endoscope and causing more serious damage.
Remove the parts, turn on the running water, clean the outer surface of the Flexible Endoscope with a soft cloth, and wash the inside of the tube cavity with a soft brush. Connect the running water, flush the inside of the tube cavity with an empty needle until it is cleaned, and drain the water in the clamp canal. If the water gun is used for washing, please note that the pressure is less than 0.2 mpa, and wipe the lens with a cotton swab.
The Flexible Endoscope is put into the enzyme solution, and the extraction enzyme solution is injected into the lumen. The soaking time is determined according to the concentration of the enzyme solution, and the soaking time is not more than 1 hour. Accessories, buttons and valves should be soaked in enzyme-containing detergent, and the ultrasonic cleaner should be turned on and shaken for 5-10 minutes.
Rinsing (final cleaning)
Turn on the deionized water, first clean the outer surface of the Flexible Endoscope with a soft cloth, and clean the inside of the tube cavity with deionized water until the inside of the tube cavity is cleaned, and the empty tube cavity after the cleaning is washed. Wipe the lens with an anhydrous alcohol swab and hold the lens with one hand.
Clean the water stains on the surface of the Flexible Endoscope with a soft cloth, and blow dry the water stains in the tube cavity of the Flexible Endoscope with an air gun, including each joint. Pay attention to the pressure of the air gun should not exceed 0.2 mpa. After blow dry, install the ETO cap for sterilization.
A. Before sterilizing the Flexible Endoscope, please completely dry and close the ETO cap;
B. Flexible Endoscope can choose gas sterilization or low-temperature plasma sterilization. Please check the table for compatibility of low-temperature plasma equipment of Flexible Endoscope.
C. Flexible Endoscope cannot be sterilized by high temperature and high pressure.